Reagent Pack, Buret, Sodium Sulfite, Iodometric, 1 mL = 1 mg
- UPC Barcode:
Description of Tests
|Analyte||System||Method/Chemistry||Standard/Equivalance or Description||Comparator||Cell|
|Sulfite (Sodium)||Buret titration||Iodometric||1 mL = 1.0 mg Na₂SO₃||NA||NA|
|Sulfite||Erythorbic acid, sulfide and ferrous iron may cause positive interference; to prevent, test for suspected interfering agent, dilute sample with DI water as necessary, and retest. Copper and nitrite may cause negative interference; to prevent, test for suspected interfering agent, dilute sample with DI water as necessary, and retest. Delay between taking sample and testing may cause negative interference; to prevent, perform test immediately after sample is taken. Quats may cause interference by forming a precipitate and causing a yellow endpoint.|
Shelf Life Concerns
REAGENT SHELF LIFE
All reagents have a shelf life, whether they are liquids, powders, crystals, tablets, or test-strip pads. If kept dry, powders and crystals are very stable; acids are also long lived. Date of manufacture is not the controlling factor when it comes to shelf life—storage conditions are more important. As with all perishables, reagents are sensitive to environmental influences and will last longer under controlled conditions.
To this end, we recommend:
- Storing reagents at a consistent temperature in the range if 36°–85°F (2°–29°C); extreme temperature fluctuation, say from a refrigerator to a hot car trunk, causes reagents to deteriorate.
- Keeping them out of prolonged direct sunlight. (Note: their brown plastic bottles help protect very light-sensitive reagents.)
- Segregating reagents from containers of treatment chemicals.
- Replacing caps immediately and tightening them carefully so that exposure to air and humidity is limited.
- Avoiding switching bottle caps, placing bottle caps on soiled surfaces, repouring reagents into contaminated containers, or touching test strip pads.
Taylor formulates its reagents to remain effective for at least one year, with only very few exceptions (molybdenum indicator in liquid form is one; after four months old it should be tested against a standard periodically). As a general precaution, replace all reagents more than one year old, or at the beginning of a new testing season.